Road & Traffic
The Basic principle of the S-sensor is that the natural frequency of a vibrating strip varies according to the tension in the strip.The heart of the S-sensor is an S-shaped metal strip fixed at both ends to a metal ring. When the ring is deformed, the natural frequency of vibration of the strip changes.
This frequency is detected as an A.C voltage in a coil positioned close to the strip. The deformation of the ring is by means of two angle brackets welded to the ring and to the stressed steelwork. The ring, the S-shaped strip and the coil are hermetically sealed inside a stainless steel housing.The forced vibration of the strip is excited by regular short pulses in the coil.
The same coil is used for detecting the frequency of the strip which is excited by regular short pulses in the coil.
|Gauge length||40 mm|
|Rated range||± 10µm (corresponding to o 50 N/mm2 at E = 200 000 N/mm2).|
|Permissible overload||10 times rated range|
|Normal sensor frequency||1600 Hz|
|Error||Less than ± 1% of rated range (linearity, hysteresis and repeatability). Error is less at lower ranges.|
|Resolution||Better than 0.01% of rated range.|
|Zero point stability||Better than ± 0.5% of rated range per month.
(At constant ambient temperature, measured 3 weeks after
mounting. Valid 5 min. after connection to the electronics).
|Temperature stability||Better than ± 0.03%/°C of rated range for a = 11.5 * 10-6/°C
(a = thermal expansion coeff. of specimen)
|Temperature range||-30°C to + 80°C.|
Need Pricing? Fill out our Request a Quote form to speed along your request.
Have a question? Ask a question to send your inquiry directly to our support team.
Contact us directly or find your nearest authorized representative.
Worldwide Representatives for Mipeg